We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||22 March 2005|
|PDF File Size:||12.55 Mb|
|ePub File Size:||20.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page Fill Program—Occurs when water level in the air separation chamber gets too low. Page 30 The Adult size comes standard with the K and can accommodate patients with upper arm circumferences of centimeters. Allergic reactions to chemical disinfectants may occur if insufficient procedures are used to remove 200k maintain the residual disinfectant at acceptable levels.
Lowering the venous pressure by reducing fresdnius blood flow rate can fressnius be effective, if using a high-permeable dialyzer. The entire Rinse program must be completed before the operator can initiate dialysis. Table 16 — The Trends Screen Buttons Button Function This button is used to display either the current or previous treatment summary data. Using A Central Venous Catheter Assembly, installation, adjustment, or repair is to be performed only by persons authorized by the facility medical director or by Fresenius USA.
K During the main program of chemical No Chemical Intake Alarm Retry manuap rinse and if problem persist, remove from service. Alarm Turn machine power off and back on. During Treatment The machine is in bypass mode during the test period.
Fresenius Medical Care K Manuals
Final Set-up Procedure Product Improvement Policy Conductivity Cells Calibration v2. If alarm is not cleared, take the machine out of service and alert a qualified technician.
Blood Leak Calibration Active Topics in this Category. If alarm is not cleared, take the machine out of service. Page Other Sodium Page TP that is to be replaced with each treatment. K High Flow Error Possible balancing vresenius problem.
Fresenius Medical Care 2008K Operator’s Manual
The Prime Amount method is a machine option that is set in the Service mode, and limits the amount of saline used in the priming procedure to a preset volume. Manually Operating the Blood Pump In the event of a power failure during treatment, the K blood pump can be manually operated to return the blood to the patient or to keep the blood in recirculation if a quick resumption of power is anticipated. A second internal TP is also installed.
Troubleshooting Alarms and Troubleshooting Page Machine alarms may not occur in every blood loss situation 5. Safety class I Degree: Section 2 — “no Water” Alarm Modules Chapter 1—Overview Modules The modules accompanying the K are located just below the 20008k panel.
Dial Valve Failure 2 Electronic self-test failure.
If the machine fails these tests and manuall cause cannot be corrected, it should not be used for treatment. Temperature Sensor Calibration tresenius Home Screen Settings Anybody check out the first paragraph on the first page of the latest PM manuals?
Voltage Detector Calibration When problems or potential problems are detected, the operator is alerted through informational messages displayed on the screen and in mqnual cases, audible alarms. Add to My Favorites. Check for residual disinfectant before using the disinfected containers.
If no time is set in the Infusion Time button and the heparin pump is turned on, it will run at the selected rate until the syringe is empty or the heparin pump is turned off. Power Failure During Dialysis Storage Preparation Before storing the K, the hydraulics should be disinfected.
Be sure the blood pump is set to the proper ID. The frresenius dotted line shows the alert level. Using The Troubleshooting Guide Rebuilding The Ultrafiltration Pump Labels And Mechanical Condition Pressing the RESET key once mutes an alarm, pressing it a second time resets the right side of the screen.
Hold for one second to turn the power off and If blood is sensed, the machine will power down with an audible alarm. Continuous Renal Replacement Therapy optional Check RO-processed water for residual disinfectant and be sure the water is within acceptable limits before connecting the water supply to the K. The machine will normally go into a Fill program when the dialyzer is first connected to the dialyzer lines, and the air within the dialyzer is being purged.
TP that is to be replaced with each treatment. Page 29 is depressed. The machine is in bypass mode during the test period. Don’t have an account? Page of Go. Recirculation And Final Set-up Procedure