Intecont Tersus – Free download as PDF File .pdf), Text File .txt) or read for weighing tasks better than % Manual and/or automatic zero setting Full. The INTECONT Tersus weighing elec- tronics is specially tion module, INTECONT Tersus optimally fits into any Manual and/or automatic zero setting. processes, the INTECONT Tersus meas- uring, control, and supervisory module, INTECONT Tersus optimally fits into any Manual and automatic zero setting.
|Published (Last):||4 February 2012|
|PDF File Size:||6.45 Mb|
|ePub File Size:||20.15 Mb|
|Price:||Free* [*Free Regsitration Required]|
Check the output regulator twrsus current limiting, speed range – Calibration error: Only personnel familiar with the device’s mode of operation may alter these parameters e. The load distribution on a single-roll weighbridge is shown by the white triangle. Any connected control systems in particular must be in safe state. An external performance level with anchor tension or speed regulation is needed for the drive in all cases.
Calibration with calibration weights: Password Confirm Defines the access protection for using the operating console.
The load cell output voltage is proportional to the platform load. When it reaches the batch setpoint, the conveyor belt and material prefeeder are turned off and the tetsus process is ended. Instead, the measured belt load is used before changing over.
Manual Intecont Tersus
The display format corresponds to Counter 1 Unit. One particular benefit is the fact that the configurable scale values are available in the CFC plan itself as binary or numerical values.
Using weighing modules with leaf spring parallel guidance, lever ratio is always 1. It can be printed out with the Batch Record function or automatically after every finished batch P Key in the final result of zero setting program and key in later in Linearization I1 parameters.
Electric W1 P This means that if the belt load changes, the measurement result for the feed rate will not accord precisely with the current feed rate at the dispatch point. Customized to meet your requirements, our comprehensive Process Advanced Service System provides you with the best service.
Otherwise, the aforementioned parameter and P Regulation stays in volumetric mode assuming the operator selected this ahead of time.
Instead, it is provided for operational shutdown. The measuring system operating company must ensure that the password is handled safely.
Preselection batching mode Preselecting the batching mode makes batching possible, although it does not initiate a batch. The effective platform length and the inclination of the belt must be measured on-site. Belt Load 3 s P The following conversion formula applies to the effective bridge length with single-roll weighbridges: The parameters values will be converted automatically if the dimensions are changed.
Pulses per length P At right angles to the belt conveyor At right angles to the belt conveyor At right rersus to the belt conveyor At right angles to the belt conveyor The angle a must be entered as the parameters. PersonnelPersonnelPersonnelPersonnel Preparation, assembly, commissioning, operation, maintenance and servicing may only be carried out by qualified personnel. This is why the control program will show the same KOR fault quotients after correcting with the value input in P A new window for entering the relative portion will open when the setpoint field is touched if the relative set-point qualifier was preselected using the operating panel.
The basic hardware equipment is aligned to the needs of the application.
Manual Intecont Tersus
Volumetric Synchronous In contrast to volumetric operation, it is not the rated belt load that is used for calculating the set belt speed. Tachometer Input A P Schenck intecont plus pdf -? Once the work is finished, they are to be applied in reverse order.
For example, setpoint specification, turning on and off, resetting the counter and acknowledging events. Special start-up and empty running functions can also be activated. Monitoring can be faulty.
The turn-off point and the utilization of “dribble feed” can be automatically adapted. Ensure that any connected peripheral devices are functioning correctly before acknowledging an event. Belt weigher manuall a constant load A non-regulated material prefeeder charges a conveyor belt. After taring, the infecont remains unchanged over one belt circuit with the empty belt. It only shows the functions that can be called up now.
The complete parameter list is printed out with the current readings using tresus Print Parameters function. Key in the sensor width in P The devices may be operated only in the provided housings. Actual Flow Rate 3 s P We will therefore list only a few important pieces of information and some recommenda-tions at this point.
The Service Values chapter explains the specific data in detail. The write protection can also be configured using EasyServe.